Amtax sc analyzer electrode maintenance kit
Availability: Order only
LCW868 accessory kit for maintenance of the Amtax sc industrial ammonium analyzer for ranges 2, 3, 4 (0.05 - 1000 mg/L NH4-N).
Includes: 3 bottles with electrolyte and 3 membranes.
For optimal performance, replace the membane cap, electrolyte and electrode regularly. The recommended service interval for the analyzer is once every six months. Some applications may require more frequent membrane and electrolyte changes.
During work with the instrument, the amount of electrolyte should be from 4 to 11 mL. With less electrolyte, the accuracy will be reduced for low measured values in the corresponding measuring range. To determine how much electrolyte is left in the electrode body, refer to the analyzer's user manual.
Note: After replacing the membrane cap and electrolyte, it may take up to 6 hours for optimum performance to be achieved.